The D-BCAM Head (A2086) is a Long-Wire Data Acquisition (LWDAQ) Device that reads out two ICX424 (or equivalent) image sensors and controls four light sources. The image sensors are mounted on two ICX424 Minimal Heads (A2076) connected to the A2086 board with two 12-way flex cables. The light sources are mounted on two Dual Laser Heads (A2074) connected to the A2086 board with two 6-way flex cables.
The D-BCAM Head is similar to the N-BCAM Head (A2083), but accommodates a second sensor with the help of pairs of the radiation-tolerant n-channel mosfet UM6K34N, and controls two more laser diodes with command bits DC12 and DC13.
Note: All our schematics and Gerber files are distributed under the GNU General Public License.A208601_Drawing.gif: Dimensions of D-BCAM circuit board.
The A2086A is a LWDAQ device of type ICX424 (6) when we read it out one pixel at a time, and device type ICX424Q (7) when we read it out with quadruple pixels. When we flash one of its LED arrays, we can use device type ICX424 (6) or ICX424Q (7), because these have the same allocation of bits for flash jobs. The A2071E and A2037E LWDAQ Drivers support both device types, but you will need firmware version 6+ for the A2071E and 18+ for the A2037E. The A2086A does not implement the functions of the enhanced ICX424 and ICX424Q device types. The table below gives the device bit allocations implemented by the A2086A.
The PXBN bit enables pixel binning to produce quadruple-sized pixels. The V1-V3 bits control the three vertical phases of the IXC424. When we assert the CCD1 bit, we select image sensor No1. Otherwise we select sensor No2. When we assert ON1-ON4 we turn on sources No1 to No4.
The A2086A does not use DC14 and DC15, which give the Virtual Device Type in the enhanced ICX424 device behavior. All four virtual devices supported by the enhanced behavior are handled the same way, as a dual image sensor, quad light source device with optional quadruple-pixel readout. When we are capturing an image of the BCAM's own lasers, we must specify that the LWDAQ operate with source device type IXC424 (6) or ICX424Q (7) as well as camera device type IXC424 (6) or ICX424Q (7). When accumulating an image in the ICX424, we must hold vertical clock phase V1 low. If we flash one of the A2086 lasers using another device type, V1 will not be held low, and the White Rectangle Error appears, like this.
The A2086 uses two pairs of UM6K34N mosfets as switches to select between the two image sensor outputs PX1 and PX2. According to our tests with x-rays, the threshold voltage of this 0.9-V mosfet drops by less than 0.1 V after 1.3 kGy. We used pairs of a similar mosfet, the UM6K31N, in the LWDAQ Multiplexer (A2085) to select from fourteen sets of LWDAQ return signals. In the A2085, the analog switches had to accommodate input voltages in the range −0.7-5.0 V. The threshold voltage of he mosfet had to be greater than 0.7 V to make sure we could disconnect a voltage of −0.7 V. The UM6K31N has a threshold voltage 1.0-2.3 V. In order to be sure to connect a voltage of +5 V with a threshold voltage of 2.3 V, we provided a 7.5 V voltage level for the gate drive of the mosfets. In the case of the A2086, the two pixel voltages are already AC-coupled onto VCOM = 1.4 V and have a maximum deviation of ±0.4 V. The UM6K34N threshold of 0.3-0.9 V permits us to connect 1.4±0.4 V with a gate voltage of 2.7 V or higher. Our logic power supply is nominally 3.5 V, and will be at least 3.1 V given +5V power is guaranteed to be 3.1 V or above, and the saturation voltage of Q2 is tens of millivolts. We can disconnect 1.4±0.4 V with a gate voltage of 0.1 V or lower. Our logic output is <100 mV for a high-impedance load like a mosfet gate. Thus we are able to make simple analog switches for our pixel voltages using pairs of these mosfets.
We use the same mosfets to switch on the laser diodes. With a gate voltage of 3.5 V, the UM6K34N has typical channel resistance less than 1.7 Ω. This resistance is insignificant compared to the 100-Ω series resistor R21 we use to drop the +15V we provide to the two A2076 dual laser heads.
he following table shows the quiescent current of several un-irradiated circuits with no sensors or sources attached.
We add the following current consumption entries to the Analyzer Tool database.
Waking up the board turns on all the ±15V power supply. The power supply switch consumes 7 mA from +15V and 3 mA from −15. The op-amps consume tens of milliamps together. When we add two image sensors (CCDs), they consume another 13 mA from +15V only.
[05-FEB-18] The A208601A printed circuit board is fully functional after correcting two errors on the schematic. First, R+ and R− were switched around on the outputs of the video amplifier outputs. Second, we must connect A to B with a wire because there is no spare logic gate in U4 to provide the connection.
[09-FEB-18] We correct A208601A and submit for fabrication as A208601C.
[15-FEB-18] We have completed the A208601B layout for the Blue D-BCAM main board and submit for fabrication.
[26-FEB-18] Firmware P2086A03 changes the pin used for signal A to match the updated Black N-BCAM layout, the black-masked A208601C, which we received last week. The circuit board now functions correctly. We measure the delay between the rising edges of A, DA, and DDA, which should be 125±25 ns.
[01-MAR-18] We prepare P2086A03.jed compiled so as to provide power-up reset for all registers, slow slew rate on all outputs, and defined 3.3-V logic levels on all outputs, and holding rather than pull-up or pull down. Our hope is that the power-up state of these ZE logic chips, when programmed in this way, will be reliable.